The optimum working volume is between 0.05 – 0.20 mL.
In the CrystalBreeder one can select the automatic evaporation option. The Evaporation tick box can only be checked if an Evaporation cap CrystalBreeder is used and a vacuum pomp is connected to the device.
The advised stirring speed is between 700 – 1000 rpm.
The turbidity measurement can be tuned to clear solutions in order to make it more sensitive. The standard procedure consists in heating your sample until everything is dissolved followed by tune and cool down to 20°C. Now you are ready to start you temperature profile for clear and cloud points determination.
The Peak Wavelength is red 645 nm.